PPT for how to calibration and verification of the contact angle meter.
According to the development and application time, the verification tools of the contact angle meter are mainly divided into three types :(1) microscope reticle, (2) standard contact angle plate of circle or Young-Laplace shape and (3) professional 3D tools for verification the contact Angle meter.
•What contact angle characterizes? The parameters for calculating contact angle？Verification or calibration?
Contact angle meter is the instrument for measuring contact angle between solid-liquid-air (or fluid), and it can also be used to measure surface tension (L/G) or interface tension (L/F). It characterizes physicochemical property of solid. So, we will also use water contact angle meter to name the contact angle meter which uses water as the probe liquid to describe the hydrophilic and hydrophobic surface of solid and the cleanness of solid surface.
Surface tension, contact angle are the two property index from contact angle meter. The measurement standard categories involved in these two indicators are: contact angle is at category 8 - geometric measurement: also known as length measurement, which is the measurement of the geometric size, position and Angle of the object; surface tension is category 10 - stoichiometry: also known as physical stoichiometry, it refers to the analysis and measurement of the composition, physical properties and basic physical constants of various substances.
So, we have two method for verifying the contact angle meter：
1, Contact angle: Measuring geometric parameters of a standard object (3D) and measuring contact angle of the same 3D object.
Then, verifying the difference between them to finish the verification of contact angle meter. The difference value of them
could be changed to a acceptable value by leveling the sample stage and view angle of lens.
2, Surface tension: Measuring surface tension of liquid with known value of surface tension and verifying the difference of
them. The verification of contact angle meter is almost same as viscometer but we cannot calibrate the value of surface
Verifying the contact angle meter, not calibrating it, because we will not and cannot adjust the parameter to meet the know value.
Tel: +1(212) 6555-777 , +1 (857) 3027458 , +1 (857) 3027459
Address: 14 Tyler Street, 3rd floor, Somerville, Boston, MA 02143